Install the driver and follow the steps in my original answer. Message 3 of And are you sure of the syntax for the measure command? Thank you Very much. How can i do with these? I followed your conversation with Joe You’ve got some code there only for the assuming the visa referecne is correct and you’re asking for the voltage not current.
|Date Added:||16 November 2005|
|File Size:||19.59 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Let’s see if I can get you on your way with the following: Go to the block diagram. Install the driver and follow the steps in my original answer.
I followed your conversation with Joe Message 4 of Then I can get the curve U t and I t.
3401a Hello Joe and Anja, Thank for your help. Here is a link for the driver: The number of bytes written is not the same as the number of bytes you want to read. You’ve got some code there only for the assuming the visa referecne is correct and you’re asking for the voltage not current.
Message 9 of First I might suggest to write in german as that might be easier for you? The same as the structure in Attachments.
How can I do this? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. You can now ask the instrument for the currently measured data i.
Before you finish your program you 3441a return your instrument from the remote state in the local state. Now I have the neu Problem. And are you sure of the syntax for the measure command?
See attached picture for additional functions: With the above named VI you first send kind of an initialization to the instrument: I don’t understand the attachment. You have to put together the building blocks to get your measurement.
A GPIB VISA TIME OUT ERRORS | Keysight Community
Message 10 of I really cannot find getting started. I never had problems with this till now. Message 3 of Now I can do this. You’re not using the 34401w for the instrument.
In this way you can communicate with every GPIB-instrument easily without the absolut need of an instrument driver. Remote operation of the instrument usually: Thank you very much for your attention. I admit you might be faster with an existing driver and of course you are more sure you didn’t forget something important but I sometimes had instruments without existing drivers.
The driver only uses the Fetch and Read commands so I can’t confirm that the Measure command is correct.
34401A GPIB VISA TIME OUT ERRORS
The front panel wi ll allow you to select the function you want to do, including DC current. Did you download the driver and install it?
Start with HPA Init.